It is always recommended to use the most recent version of software and firmware.
Let Solinst keep you up-to-date with each new software and firmware release. Register your software at: www.solinst.com/Registration/ to receive these updates automatically.
This Windows based software provides many convenient features. You can view and program AquaVent settings, begin logging sessions, monitor real-time readings, download data, manage data files, perform data adjustments, and save and export data files.
Solinst Levelogger Software is used to program all Solinst dataloggers. The software intuitively detects the type of datalogger connected, and will only display the setting options specific to that datalogger type.
2.1 System Requirements
The minimal hardware and software requirements for software installation and operation are:
|Memory: 32 MB or more||OS: Windows 7, 8 or 10|
|Display: VGA: 800 x 600 pixels, 256 colour|
|Ports: USB or RS-232 Serial Port|
Communication port setting for AquaVent communications:
|Bits per second||9600 (57,600 data download)|
2.2 Software Installation
Download the newest version of Levelogger Software by visiting:
- Insert the software CD provided.
- If the installer does not automatically start, to activate the software install click on the 'setup.exe' file located on the software CD.
- The Software Installation Wizard will guide you through the remaining installation process. Figure 2-1 shows the Software Installation Wizard.
- Restart the computer after installation is completed. Default Directory is:
Figure 2-1 Found New Hardware Wizard
2.3 Software Communication
Levelogger Software is Windows based, and is therefore used with a desktop or laptop PC. The AquaVent communicates to a laptop or desktop PC using a USB Connector Cable connected to the AquaVent Wellhead. The USB Connector Cable connects to the 10-pin (Solinst Protocol) connector on a SP or SPX Wellhead.
Figure 2-2 Communicating with Levelogger Software