Remove the storage caps from the connectors on the logger and other end of the Vented Cable.
Ensure you save all the storage caps for future use.
Line up the alignment pin in the logger connector with the alignment socket in the Vented Cable connector. Carefully push the connectors together. Twist the cable connector slightly; you will feel a small click when the properly aligned connection is made.
Tighten the stainless steel coupling while holding the stainless steel cable connector body still. Tighten the stainless steel coupling until it seats. Do not twist the Vented Cable or the logger.
Figure 1-11 Connecting a Vented Cable to the Logger
Do not twist the Vented Cable or stainless steel cable connector body when connecting to a Wellhead or AquaVent logger. Only twist the stainless steel coupling.