Ensure that the
covers the entire
sensor cell and the
logger is agitated to
(Click image to enlarge
to full size)
Retrieve the settings from the attached LTC Levelogger Junior by clicking .
Stop the LTC Levelogger Junior by clicking .
Start the Calibration Wizard by clicking the Conductivity Cal tab. The wizard will guide you through the calibration process. You can exit the wizard after any step to cancel the calibration session.
Introduction: The LTC Levelogger Junior must be connected to the computer with an Optical Reader during the entire calibration process and must be in the stopped mode. Ensure you use fresh calibration solution and allow thermal equilibration of the solution during the process. Also ensure no air bubbles are present on the sensor and that the pins are clean. See Section 10.2 for recommended methods to clean the sensor pins. See Section 10.1.5.3 for protection against biofouling conditions.
LTC Levelogger Junior Calibration Wizard Introduction
(Click image to enlarge to full size)