The Levelogger App Interface connects to the top end of a Direct Read Cable, which has a connected Levelogger at the optical end. To connect the Levelogger App Interface to the Direct Read Cable, simply hold on to the top end of the Direct Read Cable, and thread the coupling of the Levelogger App Interface onto the Direct Read Cable connector. The threaded connection is designed to be stable when installed on a Direct Read Cable in a Solinst 2" Well Cap Assembly. For more information on the Direct Read Cable and the 2” Well Cap Assembly, please see the Levelogger User Guide.
2.5.1 Levelogger Deployment
Figure 2-3 Connecting the Levelogger App Interface to a Direct Read Cable
A Direct Read to Optical Adaptor is available for connecting your Levelogger App Interface directly to a datalogger. A slip fit version is also available.
2.5.1 AquaVent Deployment
The Levelogger App Interface connects to the App Connector Cable, which is connected to an AquaVent Wellhead. To connect the Levelogger App Interface to the Connector Cable, simply hold on to the top end of the Connector Cable, and thread the coupling of the Levelogger App Interface onto the Connector Cable. The other end of the Connector Cable connects to the 10-pin (Solinst Protocol) connection on the AquaVent Wellhead. See the AquaVent User Guide for more details.
While the AquaVent is communicating with the Levelogger App, an SPX Wellhead will ignore or provide timeout responses to any requests if connected to a MODBUS or SDI-12 network at the same time.
Figure 2-4 Connecting the Levelogger App Interface to the AquaVent Wellhead