- Setup: The Levelogger Software requires the user to choose the calibration solutions. You can choose up to three solutions for a multipoint calibration of the LTC Levelogger Edge. Ensure the solutions are ready in the calibration beaker or container. Start the calibration by selecting Next.
To obtain the highest accuracy, choose the calibration solution(s) closest to your expected conductivity.
Use a two point calibration for waters between 1,413 μS/cm and 12,880 μS/cm.
Use a one point 1,413 μS/cm calibration for waters less than 1,413 μS/cm.
Use a one point 12,880 μS/ cm/80,000 μS/cm calibration for waters greater than 12,880 μS/cm /80,000 μS/cm.
Figure 6-4 LTC Levelogger Calibration Wizard Step 2