1) Getting Started
Before you can begin using your LevelSender, you need to set up the following:
- An account with a cellular service provider to receive a SIM card and setup information for your LevelSender (data usage service plan for a GSM network, including text messaging)
- Separate, dedicated email addresses for the Home Station and your LevelSender
2) Install the Batteries
A LevelSender uses three 1.5V AA user-replaceable lithium batteries. To install the batteries:
- Unscrew the stainless steel cap from the LevelSender.
- Install the three batteries with the positive ends first/down.
- Replace the stainless steel cap.
3) Connect the Antenna
- Use a Phillips screwdriver to remove the three screws from the top of the LevelSender.
- Pull off the top cap of the LevelSender.
- To attach the antenna, line up the connector and screw it onto the connection on the top of the LevelSender. (The antenna bends to accommodate well cap installation.)
4) Install the SIM Card
- Insert the SIM card into the slot in the top of the LevelSender, as shown in the photo.
- Replace the top cap (over the installed antenna), push into place and secure using the three screws.
You can leave the top cap open if you will be programming the LevelSender (see Step 8).
5) Connect Dataloggers
Each LevelSender has a single port to connect one datalogger, or a Splitter. A Splitter provides a second datalogger connection.
- Ensure the pins are lined up, and connect the Reader Cable directly to the LevelSender or Splitter.
- Connect the Reader Cable to a Direct Read Cable with a Levelogger or Barologger connected, or to an AquaVent Wellhead Connector Cable connected to an AquaVent setup.
When using a Splitter, the number 1 or 2 will identify the dataloggers in LevelSender Software. The numbers are labeled directly on the Splitter.